View and Download Hach Sc E instrument manual online. Analysis System. Sc E Measuring Instruments pdf manual download. Hach Sc Controller User Manual – Download as PDF File .pdf), Text File .txt) or read online. Hach Sc Controller User Manual . Hach Company, – .. 37 W with 25 W sensor/network card load 24 VDCpowered sc model: 24 VDC.
|Genre:||Health and Food|
|Published (Last):||21 September 2017|
|PDF File Size:||6.93 Mb|
|ePub File Size:||13.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
Warranty, liability and complaints 7.
The passcode is factory set to sc the five digits must be followed by a space to remove the trailing asterisk. Relays K1, K2, and K3. DEADBAND—Sets the range where zc100 relay remains on after the measured value decreases below the setpoint value high phase relay or increases above the setpoint value low phase relay.
The calibration history log shows the gain value, the time and date of the calibration, and the initials of the operator performing verification. Section 4 Operation Section 4 Operation 4. Index Numerics 90 Degree Detector The controller display shows the current reading plus a secondary measurement such as temperature if connected to a single sensor, or two readings with their corresponding secondary measurement readings manula two sensors are connected.
Hach-Lange SC : Owner manual
Table of Contents Section 5 Maintenance Page 23 Allow at least mm Got it, continue to print. Failure to do so could result in serious injury to the operator or damage to the equipment.
After all plumbing and electrical connections have been completed and checked, place the head on the body and supply power to the system. Disconnect and remove the existing wires between the quick connect and terminal block J5, see Figure 21 on page 4.
The turbidimeter sensor must be mounted within six feet of the controller unless an extension cable is used. Enter text from picture: Wet Verification System Startup 5. Table of Contents Add to my manuals Add.
Verify Modbus master Parity setting is set to None. Page 50 Maintenance a. Mounting Foot 2Cat. Page 51 Modbus Protocol Table 9: Close and secure the cover. Pay attention to all danger and caution statements.
The incident light Method of Detection beam is composed of a tungsten filament light source with a color temperature between and K. Twist the bulb in a counterclockwise direction, pulling out slightly, until it is released from the housing. Don’t see a manual you are looking for?
Periodic inspection of these devices is recommended to check for degradation. Page 25 Installation 4. Pipe vertical or horizontal 5. Page 26 Installation 3. The following two options are available: Please contact the Hach Service Center serving your location.
Section 2 General Information 2. This is the default setting. Failed fuses are an indication that an equipment problem could exist. Refer to Figure 29 and perform the following steps to replace the fuses: Open the controller cover.
Follow the appropriate wiring instructions per the purchased model. The warranty period for instruments is 24 months. Controller Dimension Illustrations Installation 3.
Page 39 Operation 4. Maximum cable length is 9. Comments to this Manuals Your Name. Supporting test records by Hach Company, certified compliance by Hach Company. Wear cotton gloves to protect your hands and to avoid fingerprints on the bulb.
Hach sc100 1720E Instrument Manual
Wire as shown in Table 5. Pass the cable through conduit and a conduit hub or a strain relief fitting and an available access hole in the controller enclosure. Supplied mounting hardware is shown in Figure 6, Figure 7, and Figure 8. Index A K Alarms